The HI4107 chloride ion selective electrode is a complete potentiometric cell that contains a silver/silver chloride (Ag/AgCl) reference housed within a PEI plastic body with a potassium nitrate electrolyte. The internal elements are isolated from the sample by a compressed silver chloride sensing membrane.
Hanna’s solid state combination ISEs are designed with a cone style junction. The skirted cone geometry of the sensor forms the liquid junction with the test solution at the tip of the junction cone, producing a highly stable reference potential and consistent flow rates of electrolyte into the solution. A simple press of the top portion of the ISE makes it easy to empty the electrolyte fill solution and replace when necessary.
For the HI4107 to measure chloride accurately, it is important that the ionic strength of the standards and sample be adjusted to a high and constant value with the halide ionic strength adjuster (ISA) solution. The halide ISA ensures that the activity coefficient is constant so the concentration of free chloride ions can be measured. Measuring all solutions with a constant ionic strength reduces the margin of error between measurements. The HI4107 chloride solid state ISE can measure from 50µM (1.8 mg/L) to 1M (35,500 mg/L).
Conversion Table for Cl-
moles/L (M) to ppm (mg/L)
3.500 x 104
ppm (mg/L) to M (moles/L)
2.857 x 10-5
Fixed Sensing Modules - Combination solid state ISEs are electrodes that detect free ions in a solution. The highly insoluble, compressed sensing pellet is attached to the electrode by a fixed sensing module for ease of use.
Durable PEI Body - The probe body is composed of durable polyetherimide (PEI) plastic.
BNC Connection - The HI4107 has a universal BNC connector for easy connection to any benchtop meter with a female BNC probe input.
Theory of OperationA solid state sensor electrode develops a voltage due to the ion exchange that occurs between the solution and the inorganic membrane. An equilibrium occurs due to the highly limited solubility of the membrane material in the sample. When the ionic strength of the solution is fixed by the addition of ISA, the voltage is proportional to the concentration of free ions in solution.
Description combination Measurement Range chloride (Cl-) 1.0M to 5 x 10-5M, 35,500 to 1.8 mg/L (ppm) Temperature Range 0 to 80 °C Body Material PEI Optimum pH Range pH 2 to 11 Body Length/Overall Length 120 mm Outer Diameter 12 mm Cable coaxial; 1 m (3.3’) Connector Type BNC Notes Approximate Slope -56 mV